## MSA- Measurement System Analysis

MSA is defined as an experimental and mathematical method of

determining the amount of variation that exists within a measurement process.

MSA is used to certify the measurement system for use by evaluating the

system’s accuracy, precision and stability.

### Objective of MSA:-

- To quantify the variation in measurement system.
- To ensure stability of measurement system.
- To minimize measurement system variation within

total process variation.

### Statistical Properties of Measurement System (MS):-

- Adequate discrimination (Sensitivity)

:- Least count of instrument should be 1/10^{th}of process variation

(Process variation could be specification limit) - MS variation is due to common causes, not

because of special causes. - Measurement error should be small compared to

tolerance spread/process variability.

### Measurement system error:-

- Location error [Accuracy, Bias, Stability,

Linearity] - Width Error [Precision, Repeatability,

Reproducibility, GRR]

#### Effect of MS error on measurement decision:-

- Calling OK part NG
- Calling NG part OK
- Calling common cause as special cause
- Calling special cause as common cause

#### Understanding MS errors:-

**BIAS**: – Difference between observed value and

reference value on same characteristics and same part.

**Precision**: – It is standard deviation of MS,

smaller the spread, better is precision. It is further divided into 2 parts.

- Repeatability [Within

the system variation]-Show by R chart (It is EV) - Reproducibility [Between

the system variation]- Show by average (X-bar) chart. (It is AV) - GRR [Gage repeatability & reproducibility] –

σ²(GRR)= σ²(reproducibility) + σ²(repeatability)

### MSA Acceptance criteria:-

- %R&R should be less than 10%. If more than

10 and less than 30% instrument can be accepted under some condition. But if

%R&R is more than 30% than we have to reject the instrument. - When repeatability of measurement is more than reproducibility

then maintenance/Replacement of gauge is required. - When

reproducibility is more than repeatability then training to operator is

required.

### What is ndc in MSA?

Ndc= Number of distinct categories (It should be more than

5). Calculated only in Variable MSA. In attribute MSA we calculate Kappa.

#### What are variation in MSA?

TV (Total Variation),

AV-Appraisal variation,

PV- Part variation,

EV- Equipment variation.

#### What is formula for R&R in MSA?

R&R= Sqr-root((EV²

+ AV²)

Thank You for reading and visiting on the blog. Write comment for suggestion.

You may like these articles:-

Self training program for IATF-16949

Vendor assessment System audit

**jewsfamily01@gmail.com**